Re-assign unit number for battery testers
The program does not allow any two of the testers in series to have the same unit number (Box No.). If two testers have the same Unit No., at least one could not be recognized by the software. In the advanced setup of the LANDCali, the Unit No. can be re-assigned. The Current and Voltage Ranges are determined during the manufacturing process and could not be changed.
First, connect the tester which needs to change the Unit No., with all the other testers disconnected. Close LandMon program. Then start LANDCali.exe, find “Box From” and click “Connect” (usually set it from 1 to 256, see Figure 5-5), when the tester is found and connected, click “Advanced “(see Figure 5-6). Input a new number in the “Box No” and click “Write”(see Figure 5-6). The Box No. of the device is changed.
Please note that the Current and Voltage Ranges are determined during the manufacturing process and are not supposed to be modified by customers.